A dial caliper is a versatile measuring tool that you can use to take several different physical measurements. Measuring outer and inner dimensions as well as step or hole depths is possible to an accuracy of within one-thousandth of an inch. The dials on dial calipers can move if the locking screw holding the dial in place is loosened or if the instrument is jarred. Resetting the dial on a Mitutoyo dial caliper can possibly restore accuracy to the tool if inaccurate readings are being obtained.
Things You'll Need
- Calibrated standards
Firmly close the jaws of the Mitutoyo dial caliper.
Loosen the locking screw located underneath the dial of the caliper.
Rotate the now-loosened dial so that the pointer lines up directly with the "0" marking on the dial face.
Check the accuracy of the tool with a calibrated standard. If you use a 1-inch standard, the Mitutoyo dial caliper should be accurate to within one-thousandth of 1 inch, or from 0.999 to 1.001 inches.
Tips & Warnings
- Use a calibrated standard close to the size of the object being measured to ensure maximum accuracy.
- Cease using a caliper if its readings do not match a calibrated standard's measurement.
- The gears inside a dial caliper can become damaged beyond repair, even after attempting to reset the caliper.
- Photo Credit dial caliper image by Paul Moore from Fotolia.com microprocessor image by laviniaparscuta from Fotolia.com
How to Use the Mitutoyo Digital Micrometer
In 1934, Yehan Numata created a company focused solely on the simple micrometer, aiming to develop an inexpensive, quality tool to allow...
How to Read a Dial Caliper
A dial caliper is read in hundredths of an inch, and it can be used to take outside measurements, interior measurements and...