Supplemental oxygen is an important aspect of supportive care, both in the hospital and at home. A Venturi mask -- also known as an air-entrainment mask -- is a medical oxygen delivery mask that utilizes a flow meter to deliver a precise amount of oxygen. Venturi masks are indicated for use in those individuals experiencing chronic or acute respiratory distress, or when an arterial blood gas result indicates a need.
Wash your hands with soap and warm water. Hygiene is crucial when handling oxygen masks and supplemental oxygen tubing.
Remove the mask and oxygen tubing from the packaging. Inspect the mask and tubing -- looking for cracks and discoloration. Discard any masks or tubing that are not intact.
Select the appropriate flow valve according to a physicians orders. Venturi valves are colored according to the amount of oxygen they deliver in liters per minute.
Most manufacturers follow a standard color chart. Refer to the instruction manual of the mask, if you are unsure.
Blue valves deliver 2 liters of oxygen per minute at a 24 percent concentration.
White valves deliver 4 liters per minute at a 28 percent oxygen concentration.
Yellow valves deliver 6 liters per minute at a 35 percent oxygen concentration.
Red valves deliver 8 liters per minute at a 40 percent oxygen concentration.
Green valves deliver 12 liters per minute at a 60 percent oxygen concentration.
Assemble the mask by inserting the flow valve into the mask. Attach the oxygen tubing to the mask and the supplemental oxygen source.
Place the mask on the face, so that the small pointed end of the mask rests on the bridge of the nose. The larger rounded portion of the mask should rest on the chin. Pull the elastic band gently over the back of the head to secure the mask in place.
Pull the ends of the elastic band to make adjustments to the mask, so that it fits comfortably. Make sure the mask and bands are not impeding venous blood flow. Signs of impeded blood flow are pain, numbness, and discoloration of the area.
Turn on the supplemental oxygen -- adjusting it slowly -- until you reach the desired therapy level.
Check the tubing and mask for leaks. Tighten airline connections and mask fit, as needed to fix leaks.
Tips & Warnings
- In the hospital setting, Venturi masks are for single patient use; discard additional regulators, only after the patient has been discharged.
- Photo Credit Rick Stewart/Getty Images Sport/Getty Images
Types of Oxygen Masks
An oxygen mask is used to aid in delivering oxygen into the lungs. Oxygen masks are usually made of plastic, silicone or...
What Is a Partial Rebreather Mask?
Rebreathers revolutionized the world of both scuba diving and military diving. These units are compact and light so the diver is not...
Rebreather Vs. Non-Rebreather Mask
Oxygen is vital to human life. It makes up about 21 percent of the air (the rest is nitrogen and other gases)....
How to Calculate Venturi Flow
The Venturi effect, or flow, uses the changes in pressure and velocity of fluids to determine the flow rate within pipes. This...
How to Give Oxygen to COPD Patients
Respiratory therapy techniques which are commonly used for patients with lung and pulmonary diseases such as COPD include aerosol therapy and the...
How to Calculate Oxygen Amount According to Liter Flow
If you have breathing problems, supplemental oxygen is an important part of your medical plan. Whether you are on oxygen temporarily or...
DIY Venturi Pump
Venturi pumps operate completely without moving parts except for the material moving through them. They work on the principle of conservation of...
Tracheostomy Oxygen Mask Administration Guidelines
A tracheostomy is a surgical opening through the neck and into the trachea, usually necessary after trauma, disease or surgery. As a...