A spectrophotometer is used to determine the concentration of certain compounds, such as protein, in a solution. In general, a light shines through a cuvette filled with a sample. The amount of light that is absorbed by the sample is measured. Since compounds absorb light in different spectral ranges, the right wavelength must be set for the analysis. There are numerous ways to calculate the concentration of unknown samples through spectrophotometry, however using reference standards gives the best accuracy. Furthermore, standards also indicate if the spectrophotometer is malfunctioning or other problems with the analysis.
Things You'll Need
 Spectrophotometer
 Reference standards
 Unknown samples
 Excel, spreadsheet, or graphical software
Calculating through Straight Line Equations

Analyze reference standards at the beginning of the analysis. Standards are known concentrations and are used to calibrate equipment and check accuracy. All samples should fall within the working range of the standards. If not, then samples will need to be diluted or the concentration of standards increased.

Make a scatter chart or line graph with standard concentration and absorbency readings. The concentration will be on the yaxis, the absorbency on the xaxis. For example, the standards are 1 ppm, 2.5 ppm, and 5 ppm. The absorbency given was 1 ppm= .25, 2.5 ppm= .5, and 5 ppm= .75.

Format the trend line to display an equation on the graph. The equation will show the formula y=mx+b. For example, using the standards in step 2 the equation is y= .1224x + 0.1531. Most trend lines are intercepted at zero, but it depends on the analytical method and Rsquared value.

Analyze unknown samples and record absorbency readings.

Use the equation (y=mx+b) to determine concentration of samples.
Understanding Straight Line Equation

Let "Y" equal the concentration. This is what will be solved for.

Let "X" equal the absorbency of the sample. This is the absorbency measured by the spectrophotometer.

Allow "" to equal the slope and "b" to equal the yintercept. Both are given, however if the trend line was forced through 0, then "b" will be 0.

Solve for concentration. Using the example in step 3, substitute "x" as a given absorbency of .563. Therefore:
y= .1224(.563) + 0.1531
y (concentration)= .222011
Tips & Warnings
 The units of the concentration will be the same as the standards. For example, units are parts per million (ppm) or parts per billion (ppb).
 Reference standards and samples are easily contaminated.
 Always read all parts of assay or method before developing.
References
 Photo Credit spectre angulaire image by Unclesam from Fotolia.com