A goniometer is used in the medical field to measure a joint's range of motion. It can be helpful in determining if a patient is lacking in mobility due to an injury, or how well he is recovering after sustaining an injury. The term "goniometer" comes from two Greek words that mean "angle" and "measure." This is essentially what you are doing when you measure range of motion--measuring an angle.
Place the goniometer over the fulcrum of the joint. This will vary depending on which joint you are measuring. Place the stationary arm of the goniometer along the stationary line of the body (again, this will vary depending on which joint you are measuring), and the movable arm on the moving part of the body.
Ask your patient to move the joint in the desired direction. Have the patient move to her fullest extent of motion, following the movement with the movable arm of the goniometer. Make sure the stationary arm stays straight.
Stabilize the stationary portion of the body. This is the part of the body that is proximal (closer to the midline of the body) to the joint you are testing. It is important that the patient does not move his body while moving the joint; this step isolates the joint movement for a more accurate measurement.
Look at the reading on the goniometer before removing it from the patient's body. Ensure that you take an accurate reading of the degree of motion on the goniometer, and that you consistently use the same stationary and movable landmarks on the body when measuring, to ensure consistency. Be sure to record the range of motion for the joint.
Tips & Warnings
- Stabilize the area of the body that is proximal to the joint being tested. If this area is small (such as the lower leg, if you are testing the ankle), you can hold the stationary arm of the goniometer to the proximal section of the body while stabilizing.
- Be careful not to move the stationary arm of the goniometer when taking readings. This will throw off your final reading.
- Photo Credit http://faculty.washington.edu/kepeter/119/images/goniometer.jpg
Types of Goniometers
A goniometer is an optical tool used to measure angles. Goniometers have a variety of uses in science and industry, and there...
How to Measure the Carrying Angle with a Goniometer
The elbow bends slightly when the arm is extended to allow the forearms to pass the hips while walking. This bend, called...
How to Use a Goniometer for Hand Measurement
Goniometry is a technique used to measure joint movement. A goniometer is a tool to measure the angle of movement. It consists...
How to Build a Goniometer
A goniometer is a device used for angle measurements. Its purpose is similar to that of a protractor, but the shape and...